加热时间: 20 分钟,
CE Test Report No# 991004/GERM/EMC
工作温度： 250° C (approximately 500° F),
**时间: 2-5 秒,
**槽尺寸： 20 厘米深 x 5 厘米 直径
外形尺寸：6.75'' x 5.25'' x 5.25'' (H x W x D)
Specifications: Warm Up Time: 20 min, Working Temp: 250° C (approximately 500° F), Sterilization Time: 2-5 sec, Well Size: 20 cm deep x 5 cm diameter
GERMINATOR 500 Glass Bead Sterilizer
The GERMINATOR 500 is a simple, inexpensive way to quickly decontaminate your micro-dissecting instruments between procedures. The unique stainless steel glass bead bath remains at a constant 500 degrees F, allowing you to insert the tips of your instruments whenever you wish to decontaminate them. And it’s fast...most micro-dissecting instruments decontaminate within 15 seconds! No messy chemicals and no need for open flames in the lab.
|GER-5289||Glass Bead Refill Package|
The beads will eventually, after extensive use, be worn or broken to powder, which will settle to the bottom of the well. When you notice it is difficult to push the instruments in deep enough, replace the beads.
- Features include:
- Easy to use
- Fast, effective decontamination
- Small footprint saves bench space
- Proven durable design
- Perfect for micro-dissecting scissors and forceps
- CE Mark
- 1 YEAR WARRANTY!
- Modular Cord
- Exclusive Well Collar
- Measurements: Unit: 6.75'' x 5.25'' x 5.25'' (H x W x D)
- Stainless Steel Well: 2.0'' diameter x 4.0'' deep (inside dimensions)
- Weight: 4.5 lbs.
- Electrical: 230/250 Volts 60Hz 0.85A 110W
- CE Test Report No# 991004/GERM/EMC
- Operating Temp: 500ºF 15º (233ºC)
- Glass Beads: 1.5mm lead free glass balls
Dry Sterilize Surgical Instruments in Seconds Meets NIH Guidelines
NIH guidelines recommend sterilizing instruments for small animal surgery. Hot glass bead sterilizers are ideal for this application. STAYS COOL TO THE TOUCH - The chassis stays cool, even at the end of a full day of operation. The air above the unit is cool enough to insert an instrument into the well without discomfort. Insert the surgical instrument into the well, sterilization of the inserted parts by dry heat (250° C) takes 2-5 seconds. The handles of the instruments, outside the heat, do not get hot. The metal in the beads is heated only on the surface, not to the depth, so cooling is quick after removal. Within 30 seconds, the surgical instruments are cool enough to use.